Published: June 23rd, 2016
Here we describe a procedure for studying freeze-fractured plant tissues. High-pressure frozen leaf samples are freeze-fractured and double-layer coated, yielding well preserved frozen-hydrated samples that are imaged using the cryo-scanning electron microscope at high magnifications with minimal beam damage.
Cryo-scanning electron microscopy (SEM) of freeze-fractured samples allows investigation of biological structures at near native conditions. Here, we describe a technique for studying the supramolecular organization of photosynthetic (thylakoid) membranes within leaf samples. This is achieved by high-pressure freezing of leaf tissues, freeze-fracturing, double-layer coating and finally cryo-SEM imaging. Use of the double-layer coating method allows acquiring high magnification (>100,000X) images with minimal beam damage to the frozen-hydrated samples as well as minimal charging effects. Using the described procedures we investigated the alterations in supramolecular distribution of photosystem and light-harvesting antenna protein complexes that take place during dehydration of the resurrection plant Craterostigma pumilum, in situ.
Oxygenic photosynthesis, originating in ancient cyanobacteria, was inherited by algae and land plants by endosymbiotic events that led to development of the chloroplast organelle. In all modern-day oxygenic phototrophs, photosynthetic electron transport and the generation of proton-motive force and reducing power are carried out within flattened sac-like vesicles termed 'thylakoid' membranes. These membranes house the protein complexes that carry out the light-driven reactions of photosynthesis and provide a medium for energy transduction. The thylakoid membranes of plants and (some) algae are differentiated into two distinct morphological domains: tightly app....
1. Cryo-fixation of Leaf Tissues by High-pressure Freezing
Note: This section describes how to carry out high-pressure freezing of leaf tissues for a freeze-fracture experiment. For considerations related to plant samples see29. This can be adapted for other types of tissues or samples with some modification.
Figure 1 shows cryo-SEM images of platelets containing high-pressure frozen, freeze-fractured Craterostigma pumilum leaf pieces. In some samples, large regions of fractured cells are obtained (Figure 1A). In others, the leaf piece stays tightly bound to the upper disc and is knocked off along with it (Figure 1B). However, even in the second case, some leaf tissue may remain attached to the knife grooves on the platelet (F.......
The technique described in this paper allows investigation of freeze-fractured membranes within the context of well-preserved high-pressure frozen plant tissues by cryo-scanning electron microscopy. The major advantage of using these procedures is that sample preparation is purely physical; no steps involving chemicals or dehydration are necessary. Thus, it allows studying biological structures at a near-native state26,32. The benefit of using leaf tissues is that one can obtain information on the overall cell.......
We thank Andres Kaech (University of Zurich) for his helpful advice on scanning electron microscopy imaging. This work was supported by the United States-Israel Binational Agricultural Research and Development Fund (grant no. US-4334-10, Z.R.), the Israel Science Foundation (grant no. 1034/12, Z.R.), and the Human Frontier Science Program (RGP0005/2013, Z.R.). The electron microscopy studies were conducted at the Irving and Cherna Moskowitz Center for Nano and Bio-Nano Imaging at the Weizmann Institute of Science.....
|Engineering Office M. Wohlwend GmbH, Switzerland
|Platelets are of 3-mm diameter and 0.5-mm-thick (Type A) with 0.1/0.2-mm-deep cavities (of diamater 2 mm). Similar platelets can be obtained from Leica Microsystems.
|Electron Microscopy Sciences
|Dumont (Switzerland) Durostar style #5 tweezers; Can be substituted with other high-precision tweezers.
|high-pressure freezing machine
|High-pressure freezing alternatives: 1. HPF Compact 02, Wohlwend GmbH; 2. HPM 010, RMC Boeckeler; 3. EM PACT2, Leica Microsystems; 4. EM HPM 100, Leica Microsystems; 5. EM ICE, Leica Microsystems.
|EM BAF 060
|cryo preparation loading stage
|specimen holder for univeral freeze fracturing
|Clamp holder for specimen carriers of diameter 3 mm
|vacuum cryo-transfer shuttle
|EM VCT 100
|scanning electron microscope
|cryo SEM stage
|image acquisiton software
|SmartSEM, Carl Zeiss Microscopy GmbH
|image analysis software
|Fiji/Image J, National Institute of Health
Copyright © 2024 MyJoVE Corporation. All rights reserved