Published: November 19th, 2012
We present a method of creating a thinned-skull cortical window (TSCW) in a mouse model for in vivo OCT imaging of the cerebral cortex.
Optical coherence tomography (OCT) is a biomedical imaging technique with high spatial-temporal resolution. With its minimally invasive approach OCT has been used extensively in ophthalmology, dermatology, and gastroenterology1-3. Using a thinned-skull cortical window (TSCW), we employ spectral-domain OCT (SD-OCT) modality as a tool to image the cortex in vivo. Commonly, an opened-skull has been used for neuro-imaging as it provides more versatility, however, a TSCW approach is less invasive and is an effective mean for long term imaging in neuropathology studies. Here, we present a method of creating a TSCW in a mouse model for in vivo OCT imaging of the cerebral cortex.
Since its introduction in the early 1990's, OCT has been used extensively for biological imaging of tissue structure and function2. OCT generates cross-sectional images by measuring echo time delay of backscattered light4 by implementing low coherence light source with a fiber-optic Michelson interferometer2,4. SD-OCT, also known as Fourier domain OCT (FD-OCT), was first introduced in 19955 and offers a superior imaging modality compared with traditional time domain OCT (TD-OCT). In SD-OCT, the reference arm is kept stationary resulting in a high speed and ultra high resolution image acquisition6-9.
1. Surgical Preparation
After creating a thinned window over the cerebral cortex the vasculature should now be more visually prominent (Figure 1) and will allow for a deeper imaging depth (up to 1 mm). The right cortex is thinned to approximately 55 μm as compared to a normal skull measured at 140 μm (Figure 1) and provides greater optical clarity. Further thinning to 10-15 μm is possible11 however not necessary as the use of glass cover slips and skull plates are not implemented in our expe.......
Imaging with OCT and a thinned-skull is a novel neuro-imaging technique that has only been recently investigated15, 16. In our experiments, we demonstrated the feasibility of SD-OCT imaging through a TSCW in a mouse model in vivo. From our results, the skull is thinned to approximately 55 μm and the penetration depth is obtained at approximately 1 mm with image resolution of 8 μm and 20 μm in the axial and lateral direction, respectively. In the signal intensity profile, OCT imaging through a.......
|Artificial Tears Ointment
|Church & Dwight Co., Inc.
|Sterile Alcohol Prep Pad
|Cotton Tipped Applicators
|Betadine Solution Swabstick
|Saline Solution, .9%
|High Speed Surgical Hand Drill
|Carbide Round Bur
|CompoMaster Coarse & CompoMaster Polisher
|Braintree Scientific, Inc.
Copyright © 2024 MyJoVE Corporation. All rights reserved